Measurement Point holding Upper Limit
We have a measuring point which had a upper limit, but we have removed it . System still prompts with the warning message that the reading above the upper limit. Can anyone suggest regarding this or if tehre are any notes?
Jogeswara Rao K replied
Hello S P,
I understood the problem. It is simple. You do this.
1. Go to IK02 --> Go to Addl Data.. screen (Range value screen). Just remove the Zeros also (Erase) from the Range value screen and press Enter.
2. You are out of the pop-up now. Save the measuring point.
Next time when you go to this screen it looks like this:
After this, you will not get the warning message. What is happening now is that the system is considering the 0,00 value also as a value. So, any value in the Counter Reading of IK11, is giving rise to a warning.